- It is a self rectified
X-Ray generator with Stationary Anode X-Ray Tube.
- Radiography Output
60mA, 40mA, and 25mA at 30 to 100 KVP.
- Fluoroscopy Output
0 to 3 mA (Continuously variable) at 30 to 100 KVP.
- Radiography KVP 30
to 100 KVP in steps of 5 KVP.
- Line Voltage compensation
175 to 250 Volts in steps
- Technic selection
Independent selection of Fluroscopy, Radiography.
- Bucky Radiography,
Spot Film and mA selection of 25 mA, 40mA and 60mA
- Solid state electronic
timer having range of 0.02 to 10 seconds in 24 steps.
- Built-in overload
protection for safely of X-Ray Tube.
- Indicators for Line,
Overload, X-Ray and Fuse.
- Power supply requirements
Single Phase, 230 Volts, 50Hz Ac Power supply.
- Standardisation and
- The unit conforms
to radiation protection requirements of AERB-BARC, Government